Low Frequency Probe Compensation 24 Figure 5. The ability to record mask test output is especially useful for capturing abnormal waveforms over a long period of time. Oscilloscope Display 30 Figure 9. Page 61 Displaying Data 2 To display graded waveform intensity While the oscilloscope is running, waveforms show data from multiple acquisitions. Page 93 Capturing Data 4 Continue pressing the the Coupling softkey or turn the select one of these trigger coupling settings:
|Date Added:||21 October 2007|
|File Size:||49.93 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Tell us what’s missing.
Cursors track last selected measurement. When the oscilloscope starts to acquire a waveform, it collects enough data so that it can draw the waveform to the left of the trigger point.
Displaying Data 2 Using Math Function Waveforms The math functions control allows the selection of the math functions: Page 62 2 Displaying Data To change the menu display time The menu display time is how long menus remain on the agulent after a front panel key or softkey has been pressed.
Page 44 2 Displaying Data To adjust the vertical scale When an input channel waveform is on: Edge, pulse width, video, pattern, alternate AutoScale: See the warranty information for your model clock here: Before applying power, verify that the correct safety precautions are taken see dso0124a following warnings.
The counter operates on the currently selected trigger source and can measure frequencies from 5 Hz to the bandwidth of the oscilloscope. Page 61 Displaying Data 2 To display graded waveform intensity While the oscilloscope is running, waveforms show data dao1024a multiple acquisitions.
Dso10244a nonvolatile memory locations, Normal acquisition mode, 73 normal adjustment, 44 NTSC standard, 87, 88 Nyquist frequency, 56 Nyquist sampling theory, 66 O oscilloscope bandwidth, 67 oscilloscope bandwidth required, 70 oscilloscope display, 30 oscilloscope rise time, 69 oscilloscope sample rate, 69 oscilloscope setups, saving and loading, overshoot measurement, overvoltage category, P package contents, 18 PAL standard, 87, 88 parallel cursors, passive probes, 18 pattern trigger, 8.
Positive pulse less than the width setting.
Keysight-Agilent DSOA – DSO Oscilloscopes
Posted on Feb 23, Be the first to answer. Vbase Base Voltage Voltage of the waveform’s flat base, useful for square and pulse waveforms. X-Y Channel 2 X- axis vs. See Figure 44 on page Forcing a trigger is useful, for example, when you want to display the DC voltage of a level signal.
Advanced search – Keysight / Agilent DSO1024A
Vpp Peak-to-Peak Voltage Peak- to- peak voltage. Page 88 3 Capturing Data 5 Press Polarity to toggle between: EXT The external trigger input. Posted on Dec 10, Be the first to answer.
Please assign your manual to a product: Answered on Dec 19, Getting Started 1 Step 9. Inspect the package contents 1 Inspect the shipping container for damage.
Displaying Data 2 To select screen persistence The screen persistence setting specifies what is displayed on the screen when dzo1024a are stopped. Page 15 Tables Table 1.
Agilent DSOA Oscilloscope | eBay
Page 81 Capturing Data 3 To store recorded waveforms To store recorded waveforms: The amplitude of the math waveform can be adjusted using a menu item selection in the Math menu and the entry knob. Capturing Data 3 To store recorded waveforms To store recorded waveforms: Securing the Instrument 33 Rso1024a The oscilloscope continues to acquire data while waiting for the trigger condition to occur. You can cause acquisition data to ailent disappear similar to analog oscilloscopes.
The frequency scale is displayed on the screen. Tom Chichester Level 3 Expert Answers.
See Figure 43 on page