RF, Microwave, Antennas and Optics:: I think the reason why you can’t get query the instrument during a sweep is because that sweep is a single command. Message 6 of 8. Message 11 of I’ve never had much luck with RAW resources but it’s worth a try. Message 8 of
|Date Added:||11 February 2006|
|File Size:||14.38 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 18 of The Labgiew ESG is an adaptable platform with optional capabilities to customize the instrument for baseband and RF test applications ranging from simple distortion test and general purpose troubleshooting to baseband coding algorithm development, advanced transceiver design verification, and high volume manufacturing.
Message 4 of 8. I don’t remember the exact process but you should select the manual mode for creating the resource and enter the ip address.
Message 15 of Do somebody use Agilent ec to generate any waveform by programming iqdata to download into memory.
Generating OFDM with Labview through E4438C ESG
Message 7 of 8. I have no problem controling the signal generator with all the instrument driver provided it is just that I do not know how to view the current frequency with labview, so I don’t think it is a similar problem.
I use a fair amout of Agilent equipment and this has worked extremely well for me. The wfm file is downloaded into ecand the data in the file is transmitted from beginning to the end. I now know enough to know that I have no clue about anything at all.
Generating OFDM with Labview through EC ESG | Keysight Community
Ok, found out that I need to recycle the Sig. Message 12 of It sounds like your SigGen rebooted OK. I can’t seem to find any of the VI that is able to view the current frequency the signal generator lsbview at.
C Learning new things everyday Thanks for the suggestion. Message 8 of 8. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. The drivers are bugged in my opinion.
I haven’t implemented this because it’s a bit of a change to some legacy ayilent and I haven’t had time to experiment with the implementation. Proud to say that I’ve progressed beyond knowing just enough to be dangerous. See search results instead: Message 1 of 8.
You can find more examples from NI Example Finder: Message 5 of 8. Message 19 of Message 6 of 8.
Agilent ESG Series Signal Generator
Does the returned ID really have a space after the comma, or did you just type it that way out of habit? I don’t know where to go from here. The next agilennt cannot start until the last one completes. But I do not Know how I can solve the problem.
I think the reason why you can’t get query the instrument during a sweep is because that sweep is a single command.
Yes I have look through all the examples in the instrument driver there. Message 13 of